View and Download Dixell XR20C installing and operating instructions online. Digital controller with off cycle defrost. XR20C Controller pdf manual download. View and Download Dixell XR20CX installing and operating instructions online. Digital controller with off cycle defrost. XR20CX Thermostat pdf manual. QVWDOOLQJ DQG 2SHUDWLQJ,QVWUXFWLRQV Digital controller with off cycle defrost. SET: To display target set point; in programming mode it selects a.
|Published (Last):||8 October 2017|
|PDF File Size:||2.90 Mb|
|ePub File Size:||13.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter text from picture: The instrument is fully configurable through special parameters that can be easily. The instrument shall not be used for purposes different from those described hereunder.
By pressing ‘print’ button you will print only current page. To change the Set value push the 9 or 8 arrows within 10s. Push and immediately release the SET key: Push the SET key for more than 2 seconds to change the Set point value.
In case of fault in the thermostat probe the start and stop of the compressor are timed through.
To print the manual completely, please, download it. Consider the maximum current which can be applied to each relay see Technical Data. Installing and Operating Instructions. Ensure that the wires for probes, loads and the power supply are separated and far. The “Lo” message will be displayed followed by the minimum temperature recorded.
To memorise the new set point value push the SET key again duxell wait 10s.
Each LED function is described in the following table. Do not expose to water or moisture: Don’t have an account? Page 4 Ptb Map code Dixell S. Push the DEF key for more than 2 seconds and a manual defrost will.
Ptb Map code Dixell S.
The digital input can operate as. By pressing the 9 key again or by waiting 5s the normal display will be restored.
Defrost is performed through a simple stop of the compressor. Hold press the SET key for more than 3s, while the max. In case of failure or faulty operation send the instrument back to the distributor or to. To display target set point; in programming mode it selects a parameter or confirm.
In case of applications in industrial environments, the use of mains filters our mod. To confirm the operation the “rSt” message starts blinking and the normal temperature. CCS Set point for continuous cycle: The instrument must not be. Check the supply voltage is correct before connecting the instrument.
To see the max. It provides a relay output to drive the. Comments to this Manuals Your Name.
This manual is part of the product and should be xr200c near the instrument for easy and. By pressing the 8 key again or by waiting 5s the normal display will be restored. DOWN To see the min stored temperature; in programming mode it browses the. Don’t show me this message again.